Constant current measurement load.
IL01 is a simple constant current load for testing small power supplies.
Another application is as a current limiter for testing small circuits (in case you do not have a current limited power-supply).
The circuit was originally built on strip-board. This wears out quite quickly, so I built this version on a PCB.
The strip-board circuit description is available in the download section.
The current range is 10 mA to 320 mA in 10 mA steps plus additional 50 mA, 100 mA and 200 mA settings.
The voltage range is 6 V to around 35 V within the power limitation of the circuit.
Fig.1: Basic circuit.
The LM317 maintains a voltage of Vref across Rset so the value of Rset determines the output current Iset.
J13..J20 are jumpers or DIP switches. J13..J17 allows the output current to be set from 10 mA to 320 mA. J18..J20 are single-switch settings for 50, 100 and 200 mA.
The minimum current where the LM317 is guaranteed to maintain regulation is 10 mA so this is the minimum possible value for Iset.
The current Iadj is added to Iset. This is typically 50 µA ( max 100 µA ) giving an acceptable error of up to 1%.
The major error sources in the circuit are the value of Vref ( 1.2 V to 1.3 V ) and the fact that Rset comes in discrete values.
Table 1 shows the worst case error for Iset at 10, 20, 40, 80, 90, 160 and 190 mA.
The calculations can be found in IL01C_Calc.ods which is part of the design download.
|Rset value||Vref 1.2 V to 1.3 V||Vref adjusted to 1.3 V|
|Single 5% E24 series Rset||15.3%||9.0%|
|Single 1% E24 series Rset||12.0%||5.4%|
|Single 1% E96 series Rset||6.0%||2.0%|
|Two 1% E24 series Rset in parallel||5.1%||1.2%|
|Two 1% E96 series Rset in parallel||5.1%||1.2%|
Fig.2: Reference adjustment circuit.
R11 is adjusted so the voltage Vset is 1.3 V.
The circuit introduces an offset error of around 2% at 10 mA Iset. This can be reduced to 0.5% by increasing Rset from 130 Ω to 132 Ω ( 133 Ω // 25.5 kΩ ).
You can of course adjust the output current to an exact value at one current-setting, but the overall best accuracy is obtained by setting Vset to 1.3 V.
Also bear in mind that the LM317 has a thermal error of around 0.7% over temperature that sets a limit for the accuracy you can expect.
See IL01C_Calc.ods for details.
Fig.3: Alternate reference adjustment circuit.
The primary reason for designing this circuit was to avoid the cost of the trimpot in Fig.2. A side-effect is that it gives a more accurate setting.
Close J13, leave other jumpers open ( ref. Fig.1 ).
Apply around 10 V to the input terminals.
Short R11 (jumper J12 on the PCB). Measure the voltage across R12 (TP10, TP11 on the PCB). This is Vref ( f.ex. 1.25 V ).
Remove short across R11. Measure the voltage across R12 (TP10, TP11 on the PCB). This is Vsetm ( f.ex. 1.38 V ).
Find the current through R10:
IR10 = Vref / R10 = 1.25 / 1.00E3 = 1.25E-3 ( 1.25 mA ).
Find the current through R11:
IR11 = ( Vsetm - Vref ) / R11 = ( 1.38 - 1.25 ) / 1.00E2 = 1.30E-3 ( 1.3 mA ).
Find the voltage across R11:
VR11 = Vset - Vref = 1.30 - 1.25 = 5.00E-2 ( 50 mV ).
Find the value for R11:
R11 = VR11 / IR11 = 5.00E-2 / 1.3E-3 = 3.85E1 ( 38.5 Ω ).
Find the value for R11B:
R11B = 1 / ( 1 / R11 - 1 / R11A ) = 1 / ( 1 / 3.85E1 - 1 / 1.00E2 ) = 6.25E2 ( 62.5 Ω ).
Pick nearest standard value ( 61.9 Ω ).
R11 = 1 / ( 1 / R11A + 1 / R11B ) = 1 / ( 1 / 1.00E2 + 1 / 6.19E2 ) = 3.82E1 ( 38.2 Ω ).
Find the current through R12A:
IR12A = Vset / R12A = 1.30 / 1.54E2 = 8.44E-3 ( 8.44 mA ).
Find the current through R12B:
IR12B = Iset - IR12A - IR11 = 1.00E-2 - 8.44E-3 - 1.30E-3 = 2.60E-4 ( 0.26 mA ).
R12B = Vset / IR12B = 1.30 / 2.60E-4 = 4.99E3 ( 4.99 kΩ ).
Pick nearest standard value ( 4.99 kΩ ).
There is a sheet in IL01C_Calc.ods that does this calculation.
Fig.4: IL01 schematic, 1 of 2 identical circuits.
D10..D13 is a rectifier so I don't have to worry about input polarity when using the A1, A2 terminals. With this rectifier the circuit works down to 6 V input voltage.
The A+ and A- terminals bypass the rectifier allowing for a slightly lower operational voltage.
Jumper J10 disconnects the circuit from the input terminals.
The AI+ and AI- terminals are intended for a voltmeter for current measurement. This can be bypassed by jumper J11.
D14 is reverse-voltage protection when the A+ and A- terminals are used (it will short the power-supply).
If you want reverse-voltage protection without the risk of shorting the power-supply, omit D11, D12 and replace D13 with a piece of wire.
The original circuit did not have capacitor C10 and I have seen it oscillate under certain conditions. C10 prevents that.
The jumpers J13..J20 selects the current. These can be pin-rows with jumpers or 0.3" DIP-switches.
J13..J17 sets the current from 10 mA to 320 mA in 10 mA steps.
J18..J20 are single-switch selections for 50 mA, 100 mA and 200 mA.
The current setting is accurate within 1.2% for the prototype except for the 10 mA range which is 1.8% off.
The set current is accurate to within 0.6% for voltages from 5 V to 35 V across A+, A-.
Fig.5: Photo of mounted PCB.
Download IL01C design files.
I have boards available for this project. See the PCBs page.
Download IL01A files. These are the files for the old strip-board version. Unpack the zip-file into a separate directory (links will not work).
No known issues.
This web-page, including but not limited to all text, drawings and photos, is the intellectual property of Poul Petersen, and is Copyright ©.
Reproduction or re-publication by any means whatsoever is strictly prohibited under International Copyright laws.
The author grants the reader the right to use this information for personal use only.
Any commercial use is prohibited without express written authorization from Poul Petersen.
The information is provided on an "as-is" basis and is believed to be correct, however any use of the information is your own responsibility.
This web-site may contain links to web-sites outside Poul Petersen domain ( www.poulpetersen.dk ).
Poul Petersen has no control over and assumes no responsibility for the content of any web-site outside Poul Petersen own domain.
Poul Petersen DIY index
Poul Petersen notes index
|Copyright © Poul Petersen 2017. Last update: 20191029.||Valid HTML!|